Answering All of Your Questions About VascuChek®
After surgery, we recommend wiping the transceiver or charger with a dry, soft cloth, and flushing difficult to clean areas with 60 mL of cold water two times using a 60 mL syringe.
You should also use a soft bristle brush to brush the device for one minute, paying close attention to the speakers, seams, and labels.
Then, wipe the transceiver or charger with PDI Super Sani Cloth Germicidal wipe (Quaternary/high-alcohol formula (14.85%)) for three minutes. Do not squeeze the cloth directly on the transceiver or charger.
Next, wipe the transceiver or charger with a lint-free cloth dampened with deionized water for one minute two times using a fresh cloth for each wiping session.
Finally, let the device air dry and check for residual, organic material. If present, remove it and disinfect the transceiver again.